This User’s Manual describes how to use the E5EK compact, high-function digital Nevertheless, OMRON assumes no responsibility for errors or omissions. E5EK-AA Omron Automation and Safety Controllers 1/8 DIN VAC Digital field config datasheet, inventory, & pricing. This product was manufactured at OMRON Okayama. OMRON Okayama has obtained approvals from international certification bodies for its quality system and.
|Published (Last):||5 July 2005|
|PDF File Size:||13.15 Mb|
|ePub File Size:||3.39 Mb|
|Price:||Free* [*Free Regsitration Required]|
Error 3; RSP input error With control output coolthe conditions for switching from standard control to heating and cooling control are reached when the output function is assigned at the cooling side during omro and cooling control. Of these, only the alarm assigned as the output can be used.
OMRON E5EK User Manual
SP ramp function is enabled. If the communications function is required, add on the communications unit. How- ever, note that terminal Nos.
F Meaning Calibration data is e5rk error. For details on selecting the menu display, see page Page Revision History A manual revision code appears as a suffix to the catalog number on the front cover of the manual. A nega- tive setting sets an overlap band. The process Calibration item F Dead band The dead band is set with the set point as its center.
Dead time will be measured from the time the Temperature Controller starts operating. When a potentiometer is not connected to the controller,  is dis- played on the No. Also, a lower limit value larger than the upper limit value may be set.
The set point during SP ramp is enabled only when the SP ramp function is enabled. In this mode, you can set the communications conditions, transfer output and event input parameters to match the type of option unit set in the controller. Page 66 1 Set output to maximum. For details on the setting ranges of these parameters, see chapter 5 Level 1 Mode page Setting Alarm Type 3.
Assign these output functions to control output 1, control output 2, auxiliary output 1, and auxiliary output 2. Page 0 to 99 mode AT hysteresis 0.
The display changes to [ ] 0mA calibration display. Page 7 This s5ek introduces the E5EK. J Examples of use F Set the unit No. D Setting omrron the E5EK This description assumes that the controller is operated under the fol- lowing conditions.
Setting Up 2 Draw out the internal mechanism towards you holding both sides of the front panel. For details on how to use the controller and parameter settings, see Chapters 2 onwards.
Omron E5EK-AA2-500 Processor/Controller
Failure to heed precautions can result in inju- ry to people or damage to the e5k. When replacing the output unit, use the ER.
Also, when the input type, temperature unit and scaling sensor range are changed, set point limiter is forcibly reset to the scaling sensor range. In an open status, only control output 1 is ON.
Omron E5ek-aa Digital Controller Plug in Cover (cover Only) | eBay
Page APPENDIX F Startup Conditions of DT 1 DT will start if the temperature that has been stable varies due to external disturbance and the deflection of the temperature exceeds the stable range, and then the temperature becomes stable, pro- vided that the number of maximum temperature values is less than four.
Also, related output parameters can be used only when the communications unit EAKF is added on. Operating Condition O,ron The upper limit is set for the heating side positive valueand the lower limit is set for the cooling side negative value as shown in the follow- ing figure.
First-time users should read this chapter with- out fail. E5k tracking function is enabled.
Omron E5EK-AA2 AC/DC24 Temperature Controller – Instrument Driver – National Instruments
These parameters are valid when the power is turned ON again or Function when level 0 to 2 modes are switched. HE Revision code The following table outlines the changes made to the manual during each revision. Set Point Limiter However, note that when the set point limiter is reset, the set point is forcibly changed to the upper-or lower-limit value of the set point limiter if the set point is out of the limiter range.
If the key is pressed for less than one second, the parameters are switched. Got it, continue to print. Control output heat1: Page 3 Conventions Used in This Manual J Meanings of Abbreviations Sometimes the following abbreviations are used in parameter names, figures and in text explanations.