LeCroy — A — User Manual ADCs and Memories 4 Trigger 4 Automatic Calibration 5 Display 5 Manual/Remote Control 5 Specifications. LeCroy Service Manual. Identifier LeCroy__Service_Manual. Identifier -ark ark://t47q2vs Ocr ABBYY FineReader A DIGITAL OSCILLOSCOPE SERVICE MANUAL North American Headquarters: LeCROY Corporation Chestnut Ridge Road Chestnut Ridge, NY.
|Published (Last):||26 October 2014|
|PDF File Size:||15.37 Mb|
|ePub File Size:||8.76 Mb|
|Price:||Free* [*Free Regsitration Required]|
In parallel a 1 KOhm resistor is added. Your email is included with the abuse report. A vector is generated by a composition of two X and Y linear current ramps. Possible differences will be described.
The start level of both represent the current X,Y position, and the end level represent the X,Y coordinate to go. The gain curve is the dependence of the front end variable gain on the 16 bit DAC. Nucleoelectronica SA, 05 New Zealand: Averaging – Summation and Continuous Arithmetic lecoy incl.
HP plotter responses to some RS configuration commands have been modified.
Free Service Manuals
DC to MHz typical at the probe tip. Multi-channel lectoy – All input channels can be analyzed simultaneously to allow comparison of inde- pendent signals for common frequency-domain characteristics. The only difference being that one connector is equipped with male pins and the other with female pins. It is used to capture, analyze, display and archive electrical waveforms in fields such as analog and digital engineering, automated test and measurement, telecommunication and electronics research.
When combined with ESC.
LECROY SCH Service Manual download, schematics, eeprom, repair info for electronics experts
Turn on the persistence mode, set 1 sweep, and make a single trigger. Press “More Consts” and “Phase Constants” button. Lecroy does not archive this service manual, and I have been unable to locate a copy. Scan bj Paladinmicro – paladin paladinmicro. The minimax can be set to run on read or written data by the TMSC The front-panel controls of the A have been laid out in the style of an analog oscilloscope, making it easy to use from the very first moment.
Dip switch bank 1: The following trigger classes and operational modes are available: Extrema Mode – Keeps track of time and amplitude drift by storing extreme positive and negative values, such as glitches, over a programmable number of sweeps.
The timing of subsequent triggers in sequence mode is measured with 0. Vertical Expansion – Provides vertical scale expan- sion by a factor of up to 1 0. Specifications Scan by Paladinmicro – paladin paladinmicro. Scan bj Paladinmicro — paladin paladinmicro.
LeCroy Service Manual : Free Download, Borrow, and Streaming : Internet Archive
Waveforms and setups can be stored at rates unmatched in the industry onto or Kbyte cards. Please study the diagram and figures before attempting disassembly. Graphics Density and Plot size are menu selectable. Four different window functions are applied to indicate their effect on leakage suppression and spectral resolution.
DS is data space.
You acknowledge that all comments express the views and opinions of the original author and not those of the administrator. Repeat for 940 2. Maximum number of devices 15 Maximum bus length 20 meters or 2 meters per device, whichever is less Connection star or chain Scan by Valadinmicro – paladin paladinmicro.
Most procedures in this section will allow a technician to troubleshooting down to the board level.