This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). Agilent Technologies. PSA Series Spectrum Analyzer. This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3.
|Published (Last):||16 October 2010|
|PDF File Size:||11.27 Mb|
|ePub File Size:||2.51 Mb|
|Price:||Free* [*Free Regsitration Required]|
Press System Reference press the softkey to toggle the reference on. Restores the analyzer to its factory-defined state. You must now press the NOTE You need to use a filename that does not already exist in the current directory. Front and Rear Agilebt Features This chapter gives you an overview of the front and rear panels of your analyzer.
Initial Inspection Initial Inspection Inspect the shipping container and the cushioning material for signs of stress. Make sure there is no disk in the front-panel disk drive. In this example you are loading a trace file; Index personality options not in instrument power amplifiers preamplifiers preset factory types user, creating printer setup probes AC and DC product information rear panel symbols rear panel features reference level, setting reference, turning on 10 MHz rename files RF limiters To print the manual completely, please, download it.
Power Requirements The only physical installation of your Agilent spectrum analyzer is a connection to a power source.
Agilent Technologies E4440A Getting Started Manual
You can order the service documentation for Agilent spectrum analyzers through your Agilent Sales and Service office. Page 73 Perform an Alignment. If you wish to verify this condition, remove the signal input.
In Case of Difficulty Returning an Analyzer for Service Returning an Analyzer for Service Service Tag When you are returning an analyzer to Agilent Technologies for service, fill out and attach one of the blue service tags provided at the end of this chapter.
maanual Agilent Technologies shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material. Directories are displayed as DIR and drives have nothing listed under Type:.
Power Requirements Power Requirements The only physical installation of your Agilent spectrum analyzer is a connection to a power source. The front-panel keys are dark gray, light gray, green, or white in color.
Used on Option Noise Figure personality. To see that trace 2 is in view mode, blank traces 1 and 3 by pressing. Select the type of file you want to load setup, state, trace, limits, or corrections.
File Menu Functions File Menu Functions This chapter describes how to use the functions located under the front-panel Data storage and retrieval are handled similarly to that of personal computers PCs. For example, when you press the key on the menuit will bring up its own menu of Orientation Options And Accessories Options and Mxnual This chapter lists options and accessories available for your analyzer.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
Installation And Setup Installation and Setup Viewing Catalogs and Saving Files The analyzer stores and retrieves data similarly to the way that a personal computer PC does: If your analyzer is covered by a separate maintenance agreement, please be familiar with its terms. In this example you are renaming a trace file; press 3. To access the Load menu, press 3.
Installation and Agiletn Turning on the Analyzer for the First Time If using LAN, set the IP address of the analyzer to an appropriate number for your network one that the network recognizes, but that is not yet in use: Available only at time of purchase Chapter 5 Description Provides the factory calibration test data and the standard commercial calibration certificate at initial shipment.
Surround the equipment with three to four inches of packing material and prevent the equipment from moving in the carton. To increase the accuracy of the frequency reading in the marker annotation, turn on the frequency count function. Protecting Against Electrostatic Discharge Electrostatic discharge ESD can damage or destroy electronic components the possibility of unseen damage caused by ESD is present whenever components are transported, stored, or used.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
Page 41 Improving Frequency Accuracy Creating A Directory b. Preselection reduces mixer overload from broadband signals, reduces radiation of local oscillator harmonics back to the device under test, and reduces the level of image and multiple responses displayed.
Making a Basic Measurement This chapter provides information on basic analyzer operation. If a problem persists, you may choose to: Note that the file extension is always set by the system and for this file type the extension is TRC.
This analyzer does not contain customer serviceable fuses. Item Description Data invalid indicator Status message aggilent Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger or Trace Average Display line annotation Options and Accessories Ordering Options and Accessories Ordering Options and Accessories Options and accessories help you configure the analyzer for your specific applications, and enable you to use the analyzer.
Page 53 NOTE You should notice that the trace you saved, in this example trace 3, is loaded to trace 2.