View and Download YOKOGAWA EJXA installation manual online. EJX Series and EJA-E Series Differential Pressure and Pressure Transmitters. EJXA. Manual. Yokogawa Electric Corporation. EJX Series. HART Communication Type .. This manual describes the HART protocol communica- 1 EJXA Get Yokogawa EJXA Manual. Get all Yokogawa manuals!.
|Published (Last):||26 December 2006|
|PDF File Size:||4.36 Mb|
|ePub File Size:||12.45 Mb|
|Price:||Free* [*Free Regsitration Required]|
Pay careful be removed before connecting the line. These openings must F Don’t show me this message again.
Apply a permanent mark in the check box of the selected approval type on the certification label on the transmitter to distinguish it from unused approval types.
YOKOGAWA EJX110A Installation Manual
Page 81 Unit: Ex d IIC T Vertical height between the HP process connection and the transmitter mm F Page of 57 Go. Don’t see a manual you are looking for? Add HART 7 manual. Power Supply Voltage and Load Resistance Model And Specifications Check Installation of an Explosion- the following precautions when selecting the installation Protected Instrument location. The zero point adjustment can be The local push button on the integral indicator must made with a resolution of 0.
Introduction Type Website address: Page 83 Unit: Model 8. Be sure to use the DC power Refer to each communication manual for procedures. Page 54 9. Capsule temperature sensor problem. Page 22 3. Never insert a Drain valve screwdriver or other tool between the cap and port threads to remove the cap.
Since this will cause errors in the measurement output, select the proper piping method covered with a plastic cap to keep out dust. Page 23 Location of the mark requirements of each system. This cap must be removed before connecting the piping. The following sorts of symptoms indicate that transmitter may not be operating properly. By pressing ‘print’ button you will print only current page.
Page 30 7 Preventing Freezing If there is any risk that the process fluid in the impulse piping or transmitter could freeze, use a steam jacket or heater to maintain the temperature of the fluid.
To change the process connection, the drain vent plug must be repositioned. Please do not ground the shield on the RTD side of the cable.
Yokogawa EJXA : Manual
Page 25 4. A non-hardening silicone group sealant is recommended. NOTE Ground terminals are located on the inside and outside of the terminal box. Page 26 4. Add a, b, c and d.
Yokogawa EJXA User Manual | 85 pages | Also for: EJXA, EJXA
Do not pull the cable or subject it to excessive mechanical shock. Page 55 Revision Information: To print the manual completely, please, download it.
Special Conditions sjx110a Safe Use locations: When taking delivery of an instrument, visually check them to make sure that no damage occurred during shipment. Selecting the Installation Location If the capillary tube is too long, loosely coil the 3.
Yokogawa EJX110A User Manual
Be careful not attention to the following points when routing the impulse to damage the threads when removing this cap. Follow the domestic electrical requirements as regulated in each country.
Tell us about it. Differential pressure and pressure transmitters 66 pages.
Preparation for Starting Operation Does the selfdiagnostic indicate problem location? Always turn OFF power and shut off and release pressures before disassembly. Not applicable for output signal code F.
Grounding is always required for the proper operation of transmitters. The transmitters are thoroughly tested at the factory before shipment. Absolute pressure and gauge pressure transmitters 76 pages.
Follow the domestic electrical requirements RTD, Pt Refer to General Specification for maximum working pressure of a transmitter.