This item is no longer available. Suggested replacements. UVAS sc Probe, 1 mm, with SC Controller · Get Quote. Continuous UV Results 1 – 20 of Search results for sc Amtax sc Ammonium Analyzer, mg/L NH₄-N, V, 5 m Filterprobe sc. This item is no longer available. Suggested replacements. UVAS sc Sensor, 5 mm, with SC Controller · Get Quote. Continuously protect plant treatment.
|Published (Last):||20 October 2011|
|PDF File Size:||2.81 Mb|
|ePub File Size:||13.82 Mb|
|Price:||Free* [*Free Regsitration Required]|
Measured reading based on a gain of 1.
Rinse with deionized water before calibrating. Seat insulation against connector with no bare wire exposed. The time is available only in hour military format. Remove the quick connect fitting and wires and install the threaded plug on the opening to maintain the environmental rating. Hacch Dimension Illustrations Installation 3.
The data logs store the measurement data at selected intervals. Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions.
Hach-Lange SC User Manual | 58 pages
Maximum cable length is 9. Page 72 Visit http: Twist the bulb in a counterclockwise direction, pulling out slightly, until it is released from the housing.
See Replacement Parts and Accessories on page 55 for ordering information. Drain body and clean the body and bubble trap. Table 1 Turbidimeter Specifications Nephelometric light scatter at 90 degrees relative to the incident light beam. Pass the cable through conduit and a conduit hub or a strain relief fitting and an available access hole in the controller enclosure. Turbidimeter Installation Information Installation 3. Table of Contents Add to my manuals Add. Troubleshooting Section 7 Troubleshooting Controller Display Features Operation 4.
Sc00 Troubleshooting Table 11 presents sensor warnings displayed in the Event Log, possible causes, and corrective actions. To print the manual completely, please, download it. Do not touch the new bulb with bare hands.
Enter text from picture: In addition, the passcode also regulates function selection for relay options. Problem resolution and fuse replacement should only be performed by qualified service personnel. Replacing The Lamp Assembly Index Numerics 90 Degree Detector By pressing ‘print’ button you will print only current page. To hold the output until released: Got it, continue to print.
The incident light Method of Detection beam is hac of a tungsten filament light source with a color temperature between and K.
Section 4 Operation 4. Allow at least mm Please contact the Hach Service Center serving your location.
Hach-Lange SC 100 User Manual
The frequency will depend on the nature and concentration of dissolved and suspended solids in the sample. Wet Verification System Startup 5. Wire s100 less than 18 AWG is not recommended. Index Sample Line Installation