Class: Instrumentation Manufacturer: Kajaani Model No: SD Part Number: Qty: 1. Description: Automatic Sample Valve. follow-up through a communicator or Kajaani interface via Ethernet. Three operating modes: • Manually initiated. • Collection timer 24/7. STEC SEC Mass Flow Controller. ₪. METSO KAJAANI SD TRANSMIITER USED. ₪. MKS Flow Meter (N2) – Model ALS3BV.
|Published (Last):||19 December 2008|
|PDF File Size:||5.22 Mb|
|ePub File Size:||15.61 Mb|
|Price:||Free* [*Free Regsitration Required]|
Choosing the sampling device Hose coupling Hose clamp Sample line Fig. No additional import charges at delivery! Hose couplings for 19 mm and 25 mm hoses are included in delivery. Skip to main content.
Kajaani SD-502 Automatic Sample Valve
See the picture of sampling device body on page 6. Replacing the pressure switch Add to watch list. Turn off email alerts.
Batch samples are always led from batch digester to Defibrator with metal pipes, and the line is connected to analyzer using s5002 hose and 25 mm hose couplings!
Select a valid country. Move the closing piston back and forth by pushing it manually by part Weak weld between flange and sampling device body.
The flow opening of kajwani piston opens more when the piston moves a longer distance. Ripley, Tennessee, United States. Control cable bushing Control cable terminal strip Fig.
For pulps containing chlorine compounds, when the process consistency is 0. Turn the shaft 5 by the 13 mm spanner gap to the required direction. Connect the wires to the pressure switch part 6. This ensures that no objects remain between the sealing surfaces and prevent the piston from closing completely.
The seal ring is secured in place with a press-on fit. SD installation, operating, service – 4. Item Location see all Item Location.
Piston rotation guard, pushed to the left d Continue from point 6. Then pull the bush out of its casing by turning nut A. Open the water shut-off valve. Detach the collar part 4.
מדי זרימה נוזלי והמונית: פשוט לקנות באיביי בעברית – זיפי
Chemicals Chemicals must be disposed of as specified by the currently valid Safety Data Sheets. Batch samples from batch digester to Kappa Defibrator Before removing, make sure that the nut on the tool is screwed fully down see the picture.
Selecting the correct sampling device 6. Replace the rotation guard. Cutting blade Flow opening Fig. Cabling between sampling device and analyzer. Detach the socket from the solenoid wall.
KURZ 454FTB-16-HHT Single Point Insertion Thermal Mass Flow Transmitter
Connection box, front view. Make sure that the connections are tight. For additional information, see the Global Shipping Program terms and conditions kamaani opens in a new window or tab No additional import charges on delivery Delivery: The order of the wires is not important. Metso Automation Manufacturer reserves the right to changes without prior notice.
kajaani in Business & Industrial | eBay
More detailed information is given in the manuals for each sampling device. If you detect an air flow, replace the diaphragm; see point 6. After adjustment, close the cover carefully. Slotted cheese-head screw M4x12 acid-proof DIN Sample line connectors to the analyzer. Refine your search for kajaani.